Olympus produced equipment for connecting their Olympus OM 35 mm film cameras to trinocular versions of their BH (BHA, BHB and BHC biological models, BHF fluorescence model and BHM and BHMJ metallurgical models) and CH compound microscopes and IMT inverted microscope for photomicrography. These accessories can either be adapted or replaced to allow Canon EOS, Four Thirds, Nikon, Pentax, Sigma, Sony and other digital SLR cameras to be used instead of OM cameras, with the digital sensor located in exactly the same plane as the film.
Starting from the top, you need the following 6 items:
The BH microscopes were discontinued around 1980, and so most of the items referred to on this page (including the Photomicro Adapter L, the FK photo eyepieces, the PM-ADF eyepiece adapter and the BH-TR30 and BH-TR45 trinocular heads) are also discontinued. The best places to find them are eBay and microscope dealers who sell used equipment.
You do not need a lens on the camera, because the photo eyepiece is designed to project an image directly onto the sensor. The best digital SLRs to use as replacements for an Olympus OM have a 36×24 mm full-frame sensor, for example the Canon EOS 5D Mark II and 5D Mark III, allowing you to carry on using the same FK 2.5× photo eyepiece.
A cheaper option is a digital SLR with an APS-C sensor (22.2×14.8 mm), such as the Canon EOS 7D, 40D, 50D, 600D (Rebel T3i) and 1100D (Rebel T3), but the smaller sensor results in a substantially smaller field of view.
It is possible to use digital cameras with a Four Thirds sensor (17.3×13.0 mm), but the camera’s field of view is even smaller than with an APS-C sensor.
Vibration from the camera’s mirror and shutter can be a problem when taking photographs through a microscope. Traditional solutions include electronic flash, long exposure times, mirror pre-release, and stands that support the camera independently of the microscope.
Canon EOS digital cameras from the 5D Mark II, 7D, 40D, 450D and 1000D onwards have a Live View Silent Shooting mode that makes them particularly suitable for photomicrography. In this mode, vibration is almost eliminated, because the exposure is started electronically while the mirror is already raised and the shutter is already open. This is also referred to as EFSC or electronic first shutter curtain. EOS cameras also come with EOS Utility software that allows the camera to be controlled via a USB connection from a computer, allows the computer screen to be used for checking focus and composition, and allows images to be saved directly to the computer.
For Canon EOS camera bodies, new lens adapters are readily available on eBay to enable these bodies to be fitted to the top of the Photomicro Adapter L instead of an Olympus OM camera. These adapters convert the OM bayonet on the Adapter L to an EF bayonet, and exactly compensate for the 2.0 mm difference in registration between OM and EOS bodies.
There are 3 types of adapter for using Canon EOS bodies – no chip, AF confirm chip and EMF chip. The features of the 3 adapters are:
|no chip||AF confirm||EMF|
|M (manual exposure, user sets the shutter speed)|
|Av (auto exposure, camera sets the shutter speed)|
|Spot, partial, centre-weighted and evaluative metering|
|Confirmation of manual focus by beeping and by flashing one or more of the AF points in the viewfinder|
|EXIF lens data fixed at 0 mm for focal length and F/0 for aperture|
|EXIF lens data fixed, e.g. 50 mm for focal length and F/1.4 for aperture|
|EXIF lens data can be changed by user|
For Sony NEX, Four Thirds and Micro Four Thirds camera bodies, new lens adapters are readily available on eBay to enable these bodies to be fitted to the top of the Photomicro Adapter L instead of an Olympus OM camera.
For Minolta AF/Sony α camera bodies, there are new adapters on eBay that might be suitable for use with a Photomicro Adapter L. The adapters are thick but they include a lens, so they might allow OM lenses to focus to infinity; if this is the case, then they probably are suitable.
For Canon EOS, Minolta, Nikon, Olympus E, Panasonic, Pentax, Sigma, Sony α, Sony NEX and other camera bodies, new T-2 lens adapters are readily available on eBay to enable these bodies to be used with alternative camera adapters that have a T-2 fitting on top.
Olympus produced the Photomicro Adapter L to allow their Olympus OM 35 mm film cameras to be used for photomicrography with trinocular versions of their BH compound microscopes. Lens adapters are easily available on eBay to convert the OM rear lens fitting on the top of the Adapter L so that Canon EOS, Four Thirds and other shallow-bodied cameras can be used instead of OM cameras, with the digital sensor located in exactly the same plane as the film.
The bottom of the Adapter L clamps on to the circular dovetail on an eyepiece adapter.
The Adapter L was originally finished in grey to match the BH microscope. The finish was later changed to cream to match the BH-2 microscope.
If you want to use a Nikon, Pentax or other deep-bodied digital SLR, it is not easy to convert the Adapter L, but there are at least 3 sources of alternative adapters that attach to the Olympus 38 mm circular dovetail at the bottom and take a T-2 mount at the top. I have not seen any of these adapters, so please contact the manufacturers to ensure that they are suitable.
1) Richard J. Kinch (Olympus 38mm Dovetail to T-mount Adapter) makes an adapter that maintains the proper 150.0 mm distance from the lip on which the top section of an NFK eyepiece rests to the plane of the sensor.
2) Martin Microscopes produce their MBH2T adapter; this is not listed on their website so you need to contact Martin Microscopes.
3) Diagnostic Instruments used to produce their PA1-10A adapter.
To project the image produced by the objective onto the sensor in the camera body, special photo eyepieces that sit inside the circular dovetail on the eyepiece adapter are used, not the viewing eyepieces.
The FK photo eyepieces were designed for photomicrography with the short barrel objectives that were supplied with the BH microscope, including the Plan, PlanApo, MPlan and Neo Plan ranges. These objectives have a 36.65 mm parfocal distance.
For digital SLRs with a full-frame sensor, such as the Canon EOS 5D Mark II and Mark III, the most appropriate photo eyepiece is the FK 2.5×.
For digital SLRs with an APS-C sensor, such as the Canon EOS 7D, 40D, 50D and 550D, the most appropriate photo eyepiece is also the FK 2.5×, but the field of view is substantially reduced because of the smaller sensor.
For digital SLRs with a Four Thirds sensor, the most appropriate photo eyepiece is also the FK 2.5×, but the very small sensor means that the camera’s field of view is even smaller than with an APS-C sensor.
Olympus did not make an FK photo eyepiece with a lower magnification than 2.5× that would have compensated for sensors smaller than 35 mm film.
The FK 3.3×, 5× and 6.7× photo eyepieces can be used when higher magnification (with a correspondingly smaller field of view) is required.
Olympus did not make a photo eyepiece that can cover the field of view through the BH-SWTR superwide trinocular head.
The trinocular heads for the BH microscope have a plain vertical eyepiece tube with a 25 mm external diameter
Trinocular heads have a pair of inclined tubes for binocular viewing plus a third vertical tube that takes an FK photo eyepiece.
Four trinocular heads were made for the BH microscope. They all have a plain vertical eyepiece tube and need an eyepiece adapter in order to provide the circular dovetail for attaching a camera adapter. The common trinocular head is the BH-TR45, and there is also a BH-TR30, a strain-free BH-PTR30 and an uncommon and expensive superwide head, the BH-SWTR.
On the BH-TR45, BH-TR30 and BH-PTR30 trinocular heads, both viewing eyepiece tubes are adjustable. To make the viewing eyepieces parfocal with the camera, first focus the image from a low-power objective seen through the camera. Then, without touching the microscope’s coarse or fine focus knobs, adjust the left and right viewing eyepiece tubes to bring the images seen through the eyepieces into focus.
On the BH-SWTR superwide trinocular head, both eyepiece tubes are fixed but the superwide eyepieces incorporate dioptric adjustment. To make the viewing eyepieces parfocal with the camera, first focus the image from a low-power objective seen through the camera. Then, without touching the microscope’s coarse or fine focus knobs, adjust the left and right viewing eyepieces to bring the images seen through the eyepieces into focus.
Trinocular heads for the BH-2 microscope can be used on the BH. They incorporate the circular dovetail and so the PM-ADF eyepiece adapter is not necessary.
For more information, see the following pages:
For information on how to use cameras with other Olympus microscopes, see the following pages: